Toledo Transducers calibration laboratory and calibrations are conducted according to ISO/IEC17025. Our calibration procedure 1017 is accredited by the American Association for Laboratory Accreditation (A2LA). This accreditation assures our customers that their load cells have been calibrated in conformity with national standards. ASTM recommends re-calibration of load cells once every year, and more often where heavy usage or possible damage has occurred. A company's own quality policy may also dictate a more frequent calibration schedule. Toledo Transducers is one of the country's largest labs providing certified calibrations in compression up to 2 million pounds. We can provide tension calibrations to 120,000 pounds. We make every effort to design and build the highest quality load cells. Providing high quality calibrations is an integral part of our quality system; insuring our customers the best products.
Levels of Load Cell Calibration:
ASTM E74: The highest level of calibration that we offer. It requires three runs of ten points each (in addition to 'as received' data.) Each of the three runs is taken along a different axis of the load cell. Because we gather significantly more information about the load cell, this method provides the most accurate description of the load cell's performance characteristics. As would be expected, this is also the most expensive type of calibration and is normally required only in special circumstances.
Toledo Transducers 1017: The most common type of calibration. This method consists of two runs of five data points each and also contains the "as received" data. While this provides less data than the ASTM E74 method, it is normally adequate for most uses.
Single Point Commercial: These calibrations consist of recording the maximum load twice, with a zero return between the two points. While this yields only very limited information, it is an efficient and inexpensive way of insuring that the load cell is in proper working order.
Customer Calibration: We can structure a calibration to fit the loading characteristics of your real world application. Be sure to contact us to discuss these requirements.
Please complete all fields of this form and include it with your shipment. There is a shipping label at the bottom to clip for your convenience.
Customer Action Line
Toledo Transducers at your direction can supply a recalibration due date on the new calibration certificate. Please indicate your due date and please specify the time frame.
Make your selection on the RMA form
Contact Toledo Transducers for pricing on your requirements
Instrumentation: Toledo Transducers will gladly calibrate your instrument with the load cell needing calibration. For instrumentation NOT sold by Toledo Transducers, please provide a manual. ****This will be required to prevent delays in service.****
Conditions: If Toledo Transducers, after consultation with the customer,
is unable to perform services quoted due to the load cell condition, the load cell will be
returned at the customers request. A $150.00 evaluation fee will apply.
If the load cell specifications or performance guide lines are not provided, then Toledo Transducers will proceed to report the load cells best capability.
Payment / Terms & Delivery: Be sure to indicate method of payment with the RMA to prevent any administrative delays
We accept Visa, Mastercard and American Express
A hard copy purchase order is requested
Please contact us for current delivery times
Terms are Net 30 days
F.O.B. Holland, Ohio - All domestic and international charges apply
Please advise on carrier
Accreditation and Scope
The three (3) documents listed below provide the scope and accreditation for Toledo Transducers with regards to calibration.
Scope: The range of calibrations that we are able to provide.
Accreditation: Recognizes that Toledo Transducers has passed an annual audit at 100%. All standards and calibrations are traceable to NIST.
Proficiency Testing: The standard test voluntarily performed by our laboratory and checked by an independent agency for competency.
NIST Traceability of Load Cells: NIST traceable means an accredited
laboratory's calibration standards are certified to have direct traceabilty to the NIST
Non-Accredited laboratories have NOT demonstrated "NIST Traceability" to a third party accrediting agency. Toledo Transducers accrediting agency is A2LA.
Calibration Certification # 1379.01
Document Features: The two effects that make Toledo Transducers
calibration certificates unique are the date of calibration and the serial number of
the artifact load cell being calibrated.
Notice that the "as received" data is exhibted on the certificate with the E-74 and the TT-1017 documents. Our calibration certificates enable you to track the health of your load cell thru us documenting the no load zero balance (NLZB) and the input and output impedance. In the event that the load cell has been over-loaded, these measurements can provide insight into the event serverity.
Below are sample certificates of what we provide with new Toledo Transducer load cells or those sent to us for recalibration.